NDI Equipment Settings.
Make the following initial settings on the Eddy Current Inspection Unit, NORTEC-19eII.
- 200 KHz
Null probe on test block.
Adjust phase as required to obtain horizontal lift-off.
Move probe over all three notches in test block. Adjust gain to obtain a three block vertical signal when
probe is passed over 0.040 inch notch in test block. (Refer to standard instrument display shown in Figure
Place probe on a good area in the inspection location and null. Adjust phase as required to obtain
Inspect the part.
Any signal similar to the notches in the test block is cause for rejection.
ease of accessibility and user friendliness. If the probes are changed, steps 18.104.22.168 b.
(1), (2), and (3) shall be repeated each time a change is made.
Marking and Recording of Inspection Results. Mark and record as required by paragraph 1.3.
4.18.4 Backup Method. None required.
4.18.5 System Securing. The aft landing gear, if removed, requires installation in accordance with the applicable