NDI Equipment Settings.
Make the following initial settings on the Eddy Current Inspection Unit, NORTEC-19e .
- 200 KHz
Probe drive - mid
Refer to Eddy Current Method, paragraph 1.4.11. Set up on test block as follows:
(1) Null probe on test block.
(2) Adjust phase as required to obtain horizontal lift-off.
(3) Move probe over all three notches in test block. Adjust gain to obtain a three block vertical signal when probe
is passed over 0.040-inch riotch in test block. (See the standard instrument display shown in Figure 1-7.)
Inspection Procedure. Refer to Eddy Current Method, paragraph 1.4.11 and Figure 2-24.
Place probe on a good area in the inspection location and null. Adjust phase as required to obtain horizontal lift-
Inspect the part.
Any signal similar to the notches in the test block is cause for rejection.
Either probe identified in paragraph 184.108.40.206 may be used depending primarily on the ease of accessibility and user
friendliness. If the probes are changed, steps 220.127.116.11 b. (1), (2), and (3) shall be repeated each time a change is
Marking and Recording of Inspection Results. Mark and record the inspection results as required by
Backup Method. None required.
System Securing. The tail rotor fork assembly (head and teeter stop), if removed, requires installation in
accordance with the applicable technical manuals listed in Table 1-1.