188.8.131.52 Preparation of Helicopter. The helicopter shall be prepared for safe ground maintenance and the CPG
collective stick support assembly removed in accordance with the applicable technical manuals listed in Table 1-1.
184.108.40.206 Access. Not applicable.
220.127.116.11 Preparation of Part. The part shall be thoroughly cleaned. Refer to Preparation of Part or Area for NDI,
18.104.22.168 NDI Equipment Settings.
a. Make the following initial settings on the Eddy Current Inspection Unit, NORTEC-19e
Refer to Eddy Current Method, paragraph 1.4.11. Set up on test block as follows:
(1) Null probe on test block.
(2) Adjust phase as required to obtain horizontal lift-off.
(3) Move probe over all three notches in test block. Adjust gain to obtain a three block vertical signal when probe
is passed over 0.040-inch notch in test block. (See the standard instrument display shown in Figure 1-7.)
22.214.171.124 Inspection Procedure. Refer to Eddy Current Method, paragraph 1.4.11 and Figure 6-12.
Place probe on a good area in the inspection location and null. Adjust phase as required to obtain horizontal lift-off.
Inspect the part.
Any signal similar to the notches in the test block is cause for rejection.
Either probe identified in paragraph 126.96.36.199 may be used depending primarily on the ease of
accessibility and user friendliness. If the probes are changed, steps 188.8.131.52 b. (1), (2), and (3) shall be
repeated each time a change is made.
184.108.40.206 Marking and Recording of Inspection Results. Mark and record the inspection results as required by paragraph
6.12.4 Backup Method. None required.
6.12.5 System Securing. The CPG collective stick support assembly requires installation in accordance with the
applicable technical manuals listed in Table 1-1.