2-11. TADS NSA INSTALLATION ON NIGHTSIDE TEST BENCH. (CONT)
(2) On nightside test bench (1, figure 2-15), rotate turnlock fasteners (2) 1/4 turn
counterclockwise and open door (3).
Ensure that hands and feet are clear of base assembly to prevent injury.
When system is inflated, equipment is capable of movement. Any
movement could cause injury if caution is not taken.
(3) On rear of pneumatic control panel (1, figure 2-16), flip toggle valve (2) down (on). System
should inflate and stabilize E/O test bench set. Measure gap at all four corners of base and
verify that it is 1/4-inch to 3/8-inch. If all four gaps are not equal, refer to TM 11-6625-3081-23-
HIGH PRESSURE AIR
To prevent damage to equipment, do not adjust high pressure air above
125 PSI on pneumatic control panel.
(4) On pneumatic control panel (1, figure 2-17), verify that source gage (2) indicates a maximum
of 125 PSI. Verify that the other three gages show a PSI indication. Gages will not indicate
Figure 2-15. Pneumatic Control Panel